Polygraph study hemodynamic
◦Heart rate: 20 to 300 b.p.m. ± 1 b.p.m.
◦Gains selection: 1/4; 1/2; 1; 2 and 4 cm/mV.
◦Visual and audible QRS beep.
◾3 leads patient cable: DI, DII or DIII.
◦2 Simultaneous channels:
◾5 leads patient cable: DI, DII, DIII, AVR, AVL, AVF and V.
◾10 leads patient cable: DI, DII, DIII, AVR, AVL, AVF, V1, V2, V3, V4, V5, V6.
•INVASIVE BLOOD PRESSURE
◦2 channels of invasive blood pressure.
◦2 Additional channels of invasive blood pressure (optional).
◾Systolic, diastolic and mean pressure measurement.
◾Pressure ranges selectable between automatic or manual mode, between -25 mmHg and 475 mmHg.
◾Display of systolic, diastolic and mean pressures simultaneously.
◾Edition of labels with name of selected pressure waveform:
ART: Arterial Pressure
AO: Aortic Pressure
LV: Left Ventricle Pressure
PA: Lung Artery Pressure
PAW: Pulmonary-Capillary Wedge Pressure
RV: Right Ventricle Pressure
RA: Right Atrium Pressure
LA: Left Atrium Pressure
or Edited by the user.
◾Mean pressure waveform.
Volume of injectate 3, 5 and 10 ml.
Constant K of catheter or computing constant CC.
◦Measurement of blood temperature.
BSA: body surface
CO: cardiac output
VS: systolic volume
CI: cardiac index
SI: systolic index
◦Measurement of body temperature or of the injected for
◦Temperature range: -5ºC to 45ºC.
◦Audiovisual alarm for superior limit or inferior limit of any of the
◦Heart Rate, Mean Pressure, Temperature and SpO2, FP (optional)
◦Calculation of valvular areas by 2 pressure waveforms
◦Calculation of valvular areas with 1 channel performing pullback.
◦Calculation of vascular and systemic resistance.
•CHRONOMETER: on screen for timing, example: dilatation times, etc.
•IMAGE FREEZE FOR:
◦See up to 100 pages of monitoring stored in memory.
◦Perform measurements on the frozen curves.
◦Print pages frozen at the moment.
◦Store a frozen page
•CALIPER FOR MEASUREMENT OF:
◦Time in seconds
◦ECG Amplitude in mV.
◦Invasive blood pressure in mmHg.
◦dP/dt in mmHg/sec
•MENU IN SPANISH, PORTUGUESE AND ENGLISH.
◦Use of menu through keyboard or mouse.
◦Selection of parameters without losing monitoring
• HARD DISK DATA STORAGE, WHICH ALLOWS:
◦Saving of Patient Case Stdies with patient name and study number.
◦Recover saved Patient Case Studies.
◦Delete saved Patient Case Studies.
◦Footswitch, keyboard or mouse recording command.
◦Printing of saved Patient Case studies.
◦Updating of desired parameter configuration file,
during equipment start-up.
◦Systolic, diastolic and mean pressures.
◦SpO2 and pulse rate (optional).
◦Display of 1, 8 and 24 hours.
◦Zoom in the scale of 24 hours.
◦Printing of trends.
◦Selectable between 25, 50 and 100 mm/sec.
◦Can be modified over studies in progress or saved.
•PRINTING: -Patient case studies.
◦Either fast or high definition printing modes.
◦Recording on paper during progress of monitoring.
•PULSE OXIMETRY (Optional)
◦Display of SpO2
Range: 0 - 100 %
Accuracy: 100 - 70% ± 2% adult
100 - 70% ± 3% neonate
◦Display of pulse rate
Range: 30 - 250 ± 3 p.p.m.
◦Includes one sensor.
•NON-INVASIVE BLOOD PRESSURE (Optional)
◦Start of measurement in manual mode, or automatic
mode at intervals programmable from 1 minute
to 90 minutes.
◦Patient selection ADULT or NEONATE.
◦Systolic or diastolic pressure alarms.
◦Display of the last 1440 measurements.
◦Print-out of measurements in table form.
◦This software is also included in the Electrophysiology
and Hemodynamic Poligraph System.